Belt Scale WBS-4M/12.
Belt width =1200 mm.
Flow rate = 500 t/h (Shale).
Accuracy = ± 0.25% (actual).
High accuracy measurement of material flow.
4-Rollers belt scale was delivered completely with roller station for weighing influence zone (rollers machined and balanced with 0.2mm tolerance).
Belt scale was commissioned with weighing controller Siemens BW500L.
Belt scale was testing with pre-weighted materials.
Attained accuracy is ± 0.08…0.13%.
PLACE OF INSTALLATION:
VKG OIL AS, Estonia